MEscope VES FAQ's
- We strongly recommend Windows 10/11 with the latest Service Pack installed.
- A hard disk with at least 50 Gigabytes (GB) of available space
- Microsoft DirectX 11 compatible graphics hardware
For a new installation or when updating MEscope (vt######.vtl) license, copy the vt######.vtl license file into the same folder as the MEscope installation program (MEscope-Installer.exe).
Run the MEscope installation program (MEscope-Installer.exe). It will update the license and your copy of MEscope.
If you will be installing MEscope on a computer not connected to the Internet or if an error occurs while installing the prerequisites automatically, download and install the following prerequisites:
Windows 7 Users:
Windows is no longer supported by Microsoft. As a result. MEscope is no longer supporting Windows 7. We recommend upgrading to Windows 10 for optimum performance.
Download and install Console, then connect to our Cloud Database for a demonstration of MEscopeMSS Machine Surveillance Series:
If it looks like one of these: Install HASP Driver
If it looks like of these: Install Sentinel Driver
- Download NetworkServer.zip
- Extract the files in NetworkServer.zip
- Follow the instructions in the PDF file to install the Vibrant License Server
NOTE: A MEscopeVES Software License can only be activated on one computer.
- Run MEscopeVES by executing All Programs | Vibrant Technology, Inc. | MEscopeVESfrom the Start Menu.
- MEscopeVES will attempt Internet Activation automatically and show a success message if no problems were encountered.
- If you are not connected to the Internet or Internet Activation fails, email your License ID and Activation ID to firstname.lastname@example.org. When you receive your activated license file (vt#######.vtl, where ####### is your license number), follow the instructions above to install the activated license.
NOTE: To use a cloud license, MEscope must be installed on a computer that has access to the Internet.
When prompted for a license server, cloud license users should enter email@example.com as the license server.
The Acquisition Window and Time Waveforms use the Transmissibility and Input Auto Power Spectrum. This method does not exactly match the previous method, so the magnitude of the Output APS will no longer be the same as the magnitude of the ODS-FRF. The Output APS should be slightly higher in value due to the noise contained within the signal.
MEscope ODS Videos FAQ's
Yes, you can use your own high speed camera or even a cell phone. A couple things to keep in mind.
- The FPS (Frames Per Second) of the camera must be at least twice the maximum frequency you expect to extract from the video.
- Said differently, the maximum RPM of the machine you expect to extract from the video should be less than 30 times the FPS of the camera
Software updates for the Chronos camera are free with the one year warranty. If there are any questions about how to upgrade software or if there are any concerns about the Chronos, email firstname.lastname@example.org
- Our ODS Videos software extracts sampled time waveforms (TWFs) from successive frames of a video just like TWFs are extracted from analog accelerometer signals using a multi-channel digital acquisition system
- Each pixel in each frame of a video can be considered the same as a 2D accelerometer mounted on the surface captured in the video
- All the pixel motions are simultaneously sampled in each frame, just like motions are simultaneously sampled from multiple accelerometer signals with a multi-channel digital acquisition system
- One difference between the two methods of extracting TWFs is that videos do not have an anti-aliasing filter in front of them
- Without an anti-aliasing filter, frequencies higher than the Nyquist frequency, or one half of the recording frequency (also called the FPS or frames per second), will “fold back” into the lower frequency band of the spectra
- For modal testing, using a camera that can record at a higher FPS, and/or using a shaker with a band-limited excitation signal, can help overcome the anti-aliasing problem
- TWFs extracted from a video can be processed in MEscope using signal processing methods, just like TWFs sampled from accelerometer signals are processed
- Since only outputs (structural responses) are acquired with a video, we recommend that either ODS-FRFs or Cross power spectra (XPSs) be calculated from the TWFs
- Both ODS-FRFs and Cross XPSs can be curve fit in MEscope to extract modal frequency, modal damping, and mode shapes
- But when they are displayed in animation, the mode shapes will only show deformation of the structure in the 2D plane of view of the video